Ibebot SoilQuality Sensor - Soil Moisture & Nutrient Sensor | Cloud BasedRegular price $160.00 Save $-160.00
Moisture & Nutrient Sensor
- The SoilQuality sensor is a remote sensor which connects to your phone or smart device by Bluetooth and provides real-time data of your growing media (coco, rockwool & soil).
- Soil Quality measures the Electrical Conductivity (EC) of soil and growing media.
- At a given temperature, the EC of any growing media is relative to the amount of nutrients - an increase in nutrient produces an increase in Soil Electrical Conductivity. Temperature is a factor that which affects the EC level. The higher the temperature the higher the EC. Most EC agricultural standards are given at 25°C. As a rule of thumb, the EC of water increases by 2 to 3% for each increase of 1°C.
- To capture nutrients in the growing media, a plant needs water so minerals can be dissolved in water and be available as nutrients to the plant. In a dry growing medium a plant will not be able to grow and capture any nutrients. EC is also affected by the soil moisture: when the soil is dry, the EC greatly drops even if there’s minerals available.
- Comparison of Nutrient level in time or between different samples should be done based on the same moisture level and temperature.
- The 'SoilQuality' moisture measurement is given in Absolute Volume Ratio (AVR), which is the ratio between the volume of water divided by total volume of Soil + Water after mixing both.
- At 70% AVR moisture the soil becomes extremely close to a liquid and not adequate for growing. SoilQuality provides a AVR Soil Moisture measurement between 0 to 70% (anything above 70% will still show as 70%).
- ‘Soil Sensing Probe’ should be gently cleaned with disinfectant alcohol if moisture reading doesn’t reach 70% in water.
EC Measurement in water or buffer solution
Although SoilQuality has been designed to measure EC of moist soil, it can also measure the hydroponics nutrient solutions, however the following precautions should be followed:
- Protective cap should be removed.
- During measurement, the moisture level will be saturated at 70% - this is normal.
- The ‘Soil Sensing Probe’ should be fully immersed into the solution - half immersed Sensor Probe will result in a half reading.
- The measured EC value should be compared with a reference EC value of a buffer solution given at the same temperature as the sampled liquid. Unlike most sensors in the Holland Forge range, this unit does not have ATC ( automatic temperature compensation).
- The ‘Soil Sensing Probe’ should be gently cleaned with disinfectant alcohol if EC reading doesn’t match with expected value
- Data logging, Cloud Connected & Alerts
- Soil Moisture & Soil Nutrient
- Capture your plant's activity from pot into App.
- Sensoring the perfect nutrient balance blend for plant's soil quality.
- Optimum harvest with controlled light exposure.